Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||13 September 2011|
|PDF File Size:||9.75 Mb|
|ePub File Size:||6.21 Mb|
|Price:||Free* [*Free Regsitration Required]|
Thanks for your help guys.
I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to something different.
However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at. Message 4 of 6.
Message 5 of 6. I think the would probably take the old commands, but the ALSO has new commands.
Agilent / HP 4191A RF Impedance Analyzer
msnual I am able to communicate with the device through the GPIB interface, and am also able to retrieve information. Most Active Software Boards: Matt – you’re right, the commands for the A are completely different from the – probably because of its age.
It’s 84MB, so I hope you have a fast link. Guess you never know until you try! I have worked with the A and, as I recall, the commands are nothing up4191a any other instrument I have seen.
HP A for Sale|Impedance Analyzers|Impedance Measurement Products|Test Equipment Center
Message 1 of 6. Would hp1491a mind posting the code you used on the A? IIRC, the sweep parameters were something like this: Message 6 of 6. Should I be looking at doing it with LabView or something else?
In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:.
The forums emoticons manusl over for a bit there. Sorry, that last line of code there is supposed to read: Accepted by topic author Matt C.
4191A RF IMPEDANCE ANL
Good luck with it In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows: When I gave it a try, though, it worked fine.
Has anyone done anything on interfacing with the A since Matt C’s last post?
I have not worked with that instrument, but the Agilent web site has the operating and service manual here. If not, what would creating a driver for it entail?
Keysight / Agilent A – In Stock, We Buy Sell Repair, Price Quote
Message 3 of 6. I suspect that this difference might be causing my problems. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Message 2 of 6. In manusl code above, MHz is an arbitrary frequency. You really need the manual.