This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). Agilent Technologies. PSA Series Spectrum Analyzer. This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3.
|Published (Last):||9 January 2005|
|PDF File Size:||6.22 Mb|
|ePub File Size:||17.48 Mb|
|Price:||Free* [*Free Regsitration Required]|
Don’t have an account?
AGILENT TECHNOLOGIES EA GETTING STARTED MANUAL Pdf Download.
By pressing ‘print’ button you will print only current page. Viewing Traces 1 and 2 7.
The following six models are available: Directories are displayed as DIR and drives have nothing listed under Type:. To increase the accuracy of the frequency reading in the marker annotation, turn on the frequency count function. Page 73 Perform an Alignment. A pound uncoupled function. In Case of Difficulty Returning an Analyzer for Service Returning an Analyzer for Manyal Service Tag When you are agipent an analyzer to Agilent Technologies for service, fill out and attach one of the blue service tags provided at the end of this chapter.
Agilent handheld spectrum analyzer user’s guide pages. This symbol is used manuxl indicate the ON position of the power button. Page 44 Viewing Catalogs and Saving Files The analyzer stores and retrieves data similarly to the way that a personal computer PC does: Power Requirements Power Requirements The only physical installation of your Agilent spectrum analyzer is a connection to a power source. For example, when you press the key on the menuit will bring up its own menu of Orientation EaAgilenfEaEa.
The front-panel keys are dark gray, light gray, green, or white in color. Installation and Setup Turning on the Analyzer for the First Time If using LAN, set the IP address of the analyzer to an appropriate number for your network one that the network manal, but that is not yet in use: The battery is located on the CPU board.
When you have completed e4440s printer setup manula the analyzer, press the front panel key. Using The Front Panel Using the Front Panel Entering Data When setting measurement parameters, there are several ways to enter or modify the value of the active function: Page of 80 Go.
In Case of Difficulty Before calling Agilent Technologies, however or returning an analyzer for serviceperform the quick checks listed in the problem.
To be implemented in future firmware upgrade. Restores the analyzer to its factory-defined state.
Item Name Access features used with all analyzer modes and affect the state of the entire spectrum System Keys analyzer. Viewing Catalogs and Saving Files The analyzer stores agilenr retrieves data similarly to the way that a personal computer PC does: Notice The information contained in this document is subject to change without notice.
Improving Frequency Accuracy Viewing A Signal Viewing a Signal 1. Is the analyzer turned on?
EA 3 Hz – This analyzer does not contain customer serviceable fuses. You must now press the NOTE You need to use a filename that does not already exist in the current directory. Note that the file extension is always set by the system and for this file type the extension is TRC. Viewing a Signal 1.
E4440A PSA Spectrum Analyzer, 3 Hz to 26.5 GHz
Firmware Revision Firmware Revision To view the firmware revision of your analyzer, press call Agilent Technologies regarding your analyzer, it is helpful to have this revision and the analyzer serial number available. Select the type of file you want to rename setup, state, trace, limits, screens, or corrections. This manual provides documentation for the following instruments:.
While not all of the zeros following the decimal in the active function block are significant, the numbers after the decimal in the marker annotation upper-right corner of screen are significant. Protecting Against Electrostatic Discharge Protecting Against Electrostatic Discharge Electrostatic discharge ESD can damage or destroy electronic components the possibility of unseen damage caused by ESD is present whenever components are transported, stored, or used.
Retain the shipping materials for future use, as you may wish to ship the analyzer to another location or to Agilent Technologies for service. The only green key is the key, which performs an analyzer reset A summary of all front Preset